Standard Specification for Reference Masses and Devices for Determining Mass and Volume for Use in the Physical Testing of Hydraulic Cements
This specification covers the minimum requirements for scales, balances, reference masses, and glass graduates used in the physical testing of hydraulic cements. Requirements for analytical reference masses and balances are not included in this specification. Reference masses for use in the methods of physical testing of hydraulic cement and related and similar materials shall conform at least to the requirements of Class 6 reference masses given in this specification. The maintenance tolerances given in this specification for Class 6 reference masses are to be considered acceptance tolerances for purposes of this specification and the maintenance tolerances twice these values. Requirements for scales, balances, and glass graduates shall conform at least to the requirements for capacity, precision and accuracy, readability, and sensitivity.
1.1 This specification covers the minimum requirements for scales, balances, reference masses, and glass graduates used in the physical testing of hydraulic cements.
1.2 Requirements for analytical reference masses and balances are not included in this specification, but are to be found in Test Methods C114. The use of restrictive terminology, classes, ranges, and so forth has been intentionally avoided to allow the use of this specification by other standards-writing bodies with similar requirements for reference masses and devices for determining mass, if desired.
1.3 These requirements are not sufficiently descriptive to be used as the sole specifications for the purchase of reference masses or devices for determining mass without amplification.
1.4 Values in SI units shall be obtained by measurement in SI units or by appropriate conversion, using the Rules for Conversion and Rounding given in IEEE/ASTM SI–10, of measurement made in other units.
Formato físico y digital
Nota: Precios sin IVA ni gastos de envío
Añadir a la cesta