Standard Test Method for Rubber Property-Resilience Using Schob Type Rebound Pendulum
The Schob Type rebound pendulum is designed to measure the percentage resilience of a rubber compound as an indication of hysteretic energy loss that can also be defined by the relationship between storage modulus and loss modulus. The percent rebound measured is inversely proportional to the hysteretic loss.
4.1.1 Percentage resilience or rebound resilience are commonly used in quality control testing of polymers and compounding chemicals.
4.1.2 Rebound resilience is determined by a freely falling pendulum hammer that is dropped from a given height that impacts a test specimen and imparts to it a certain amount of energy. A portion of that energy is returned by the specimen to the pendulum and may be measured by the extent to which the pendulum rebounds, whereby the restoring force is determined by gravity.
22.214.171.124 Since the energy of the pendulum is proportional to the vertical component of the displacement of the pendulum, it may be expressed as 1 – cos (of the angle of displacement) and percentage rebound resilience. RB, commonly called percentage rebound, is determined from the equation:
126.96.36.199 The rebound resilience may be calculated as:
188.8.131.52 The rebound resilience may also be determined by the measurement of the angle of rebound α. From the rebound angle α, the rebound resilience in percent is obtained according to the following formula:
1.1 This test method covers a means of determining the resilience of rubber, within a range of impact strain and strain rate, by means of the impacting and measuring apparatus conforming to the requirements described in this test method.
1.2 This test method is applicable to thermoset rubbers and thermoplastic elastomers, the hardness of which, at the specified test temperatures, lies between 30 and 85 IRHD (see Test Method D 1415) or A/30 and A/85 (see Test Method D 2240). It may also be applicable to some polyester, polyether foam, and plastic foam materials.
1.3 All materials, instruments, or equipment used for the determination of mass, force, or dimension shall have traceability to the National Institute for Standards and Technology, or other internationally recognized organization parallel in nature.
1.4 The values stated in SI units are to be regarded as standard. The values given in parentheses are for information only.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Formato físico y digital
Nota: Precios sin IVA ni gastos de envío
Añadir a la cesta